Modification of Electrode in Dermatosensory Evoked Potential
Keywords:Somatosensory, evoked potential, electrode, electrical charge.
Somatosensory evoked potential (SSEP) usually done using bipolar surface stimulator of peripheral nerve
and recorded by cortical surface or subdermal needle electrodes, in this study we modify an adhesive large
surface area, surface stimulator using the same montage and method then compare the result between them. We
took nineteen patients of median nerve SSEP, locate and identify the shape of dermatom assigned according
to text using large square adhesive surface electrode. The same as the shape of dermatom, this is the cathode
electrode, for the anode, a strip has the same length of the cathode with a width of two centimetres placed
beneath the cathode, the circular ground band between stimulated and recording electrode. The subjects
In this study prepared for routine SSEP study the result obtained of both study compared using statistical
method and we found there was a significant different between two method and the modified as higher
amplitude using lesser stimulated threshold but not significant different between latency. That’s happened
because we increased the surface area of stimulation which means more the motor nerve was stimulated.
How to Cite
- The journal allows readers to read, download, copy, distribute, print, search, or link to the full texts of its articles and allow readers to use them for any other lawful purpose.
- The journal allows the author(s) to hold the copyright without restrictions.
- The journal allows the author(s) to retain publishing rights without restrictions